Liquid Controls positive displacement (PD) flowmeters are precision liquid measurement instruments that maintain a lifetime of precise and accurate fluid metering.
The meter housing (1) is designed with three cylindrical bores (2). Three rotors, the blocking rotor (3) and two displacement rotors (4 & 5), turn in synchronized relationship within the bores. The three rotors are supported by bearing plates (6 & 7). The ends of the rotors protrude through the bearing plates where the blocking rotor gear (8) is placed on the end of the blocking rotor and the displacement rotor gears (9 & 10) are placed on the ends of the displacement rotors. These gears create the synchronized timed relationship between the three rotors.
As fluid moves through the meter housing, the rotor assembly turns and the liquid is divided into uniform sections by the turning rotors. While measured fluid is displaced by additional fluid entering the meter, the new portion of the fluid is partitioned and measured, and the fluid ahead of it is displaced out of the meter. Since the volume of the bores is known and the same amount of fluid passes through the meter during each revolution of the blocking rotor, the exact volume of liquid that has passed through the meter can be determined with a high degree of accuracy.
Because the product is separated by the capillary seal, no metal-to-metal contact is required within the metering element. This means no wear, and no wear means no increase in slippage, which means no deterioration in accuracy.
The product flowing through the meter exerts a dynamic force that is at right angles to the faces of the displacement rotors and the rotor shafts are always in a horizontal plane. These two facts result in no axial thrust. Therefore, LC meters do not need thrust washers or thrust bearings, and the rotors automatically seek the center of the stream between the two bearing plates – eliminating wear between the ends of the rotors and the bearing plates.
Source: LC webpage